Facility name: UCF Lake Nona Medical Center
Location: TALLAHASSEE, FL
Project classification: Hospital – New
Project type: Acute Care
Project number: 2689500001
Lesson Learned number: 75
Impact: Cost, Schedule, Quality
Lesson Learned/Best Practice
Four inch vent stacks were noted above third floor toilets in area C on labor and delivery BLOX-made bathrooms, however walls of BLOX bathrooms are framed with 3 5/8″ studs that could not accommodate a 4″ pipe.
Suggested Actions for Future Projects
Review vendor drawings against existing conditions or other design drawings to verify coordination of connections and dimensions.